Contacts: Jacky Jiang
FAX: 0086 371 55951125
ADD: 51 Xiasha 22nd Street, Hangzhou, China
How to sterilize masks with eto sterilizer
Technical characteristics of ethylene oxide sterilizer
It is used for sterilization of heat and damp heat sensitive articles and all articles that cannot be sterilized by high pressure. The following conditions shall be satisfied for the ethylene oxide sterilizer, such as water, electricity, steam (ethylene oxide gas, compressed air) before the sterilization procedure. Ethylene oxide has a good bactericidal power to all kinds of microorganisms, and it has a strong penetration and little harm to articles. Ethylene oxide, also known as "ethylene oxide", is a colorless and transparent liquid with the smell of ether. Gaseous ethylene oxide is inflammable and explosive, and it should be kept in special ampoules or pressure metal cans.
The ethylene oxide sterilizer shall be installed in a sterile environment close to the compressed air. In this environment, exhaust fan should be set to remove the ethylene oxide gas dispersed in the air in time. Articles sterilized with ethylene oxide shall be packed with special standard paper and plastic. The instruction card is placed in the bag, and the sterilization date is generally 1-2 years. For example, it is necessary to ensure dry bath for disinfection of all kinds of articles, especially for all kinds of pipes, so as to prevent that the sterilization effect cannot be achieved due to the presence of water. It is strictly prohibited to use ethylene oxide sterilizer for sterilization of oil articles, because the intention of sterilization cannot be achieved.
Special sterilization method of ethylene oxide sterilizer
- Previous：Inspection of mask after sterilization with EO
- Next：ETO Sterilization Process Using ETO Sterilizer Machine